Price Sheet for Temperature Controllers
Our rewards are lower costs,dynamic profits team,specialised QC,potent factories,high-quality services for Price Sheet for Temperature Controllers, We are one with the largest 100% manufacturers in China. Several large trading businesses import merchandise from us, so we could supply you with the ideal value with all the same high quality if you are fascinated in us.
Our rewards are lower costs,dynamic profits team,specialised QC,potent factories,high-quality services for Temperature Controllers, Due to the changing trends in this field, we involve ourselves into merchandise trade with dedicated efforts and managerial excellence. We maintain timely delivery schedules, innovative designs, quality and transparency for our customers. Our moto is to deliver quality solutions within stipulated time.
A .Technical specification:
1、Input: K Measurement precision: ≤1.5%FS±1B
2、Measure and control range: 0~399℃
3、Control period: 30±10Sec; Proportional band range:2～6%
4、Deviation display range: -50～50℃
5、Output contact capacity: AC220V 5A（resistance load）
6、Power voltage: AC110～220V 50Hz/60Hz
7、Working environment: Temperature: 0~50℃;
relative humidity: ＜ 85%RH, without corrode gas.
8、Overall size and installation hole（mm）： 48×96×110、44×92；
B. Connection Scheme(consult):
C. Method to use:
Connect the power and the sensor according to the connection scheme, then it displays the actual measured temperature.
The terminals of ‘1’, ‘2’ are input of thermocouple signal, the terminals of ‘6’, ‘7’, ‘8’ are relay control output. When the green light of “OUT” brighten, terminal ‘7’ and ‘8’ are connected and terminal ‘6’ and ‘8’ are disconnected. When the green light of “OUT” goes out, terminal ‘7’ and ‘8’ are disconnected and terminal ‘6’ and ‘8’ are connected. The terminal of ‘3’, ‘5’ are power supply (AC110～240V).
When the actual measured temperature don’t enter the proportional band and the setting temperature is higher than the actual measured temperature, the terminal ‘7’and ‘8’connect, terminal ‘8’ and ‘6’ disconnect. The load starts heating at this time and its temperature is going up.
When the actual measured temperature don’t enter the proportional band and the setting temperature is lower than the actual measured temperature, the terminal ‘7’ and ‘8’ disconnect, terminal ‘8’ and ‘6’ connect. The load does not heat at this time and its temperature is going down.
After the actual measured temperature entry into the proportional band, the relay start opening and closing according to the controlling rule. The higher temperature, the shorter connecting time of relay between terminal ‘7’ and ‘8’. Vice versa .The controller controls the temperature by the way of changing the average heating power of load.
Note: Time proportion adjustment: the time proportion is also known as intermittent ratio adjustment, change the average heating power to achieve control in the unit time (cycle). When the actual temperature access to the dashboard proportional band, relay release cyclically, suction, by changing the absorption and release time proportion to change the average heating power of heat load, thus changing temperature for purpose.
Note: Our company will continue to improve product technology, design specification. If change, please subject to the material object, without notice.