OEM/ODM China Digital Pid Temperature Controller Thermostat With Flush Mount
From the past few years, our firm absorbed and digested sophisticated technologies equally at home and abroad. Meanwhile, our organization staffs a group of experts devoted into the growth of OEM/ODM China Digital Pid Temperature Controller Thermostat With Flush Mount, We consider excellent as the foundation of our results. Thus, we focus on the manufacture of your best high quality items. A strict excellent management system has been created to make sure the standard of the items.
From the past few years, our firm absorbed and digested sophisticated technologies equally at home and abroad. Meanwhile, our organization staffs a group of experts devoted into the growth of Flush Mount Temperature Controller, Pt100 Temperature Controller, Thermostat With Flush Mount, We can meet the various needs of customers at home and abroad. We welcome new and old customers to come to consult & negotiate with us. Your satisfaction is our motivation! Allow us to work together to write a brilliant new chapter!
XMT*JK two-way series temperature controller
XMT*JK series two-way temperature controller can connect two sensors synchronously and conveniently, diminish the instrument’s volume. It has a separate auto-tune mode and PID parameter function with more precision and more reliability in whole machine control.
Ⅱ、Primary technical standard:
1、Input signal: CU50（-50.0～150.0℃）、Pt100（-19.9～600.0℃）、K（0～1300℃）、
3、Output method: relay AC220V 3A(resistance load)
4、control mode: difference control-on/off and PID control
5、Input power: AC85~242V 50/60Hz
6、Environment Temperature: 0 to 50℃, Humidity: ≤85%RH
7、Face Size: 160×80×110 mm Hole Size : 152×76 mm
96×96×110 mm Hole Size : 92×92 mm
72×72×110 mm Hole Size :68×68 mm
Ⅲ、Panel board instruction(consult)
Ⅴ、Code setting mode
1.Accurately connect uires according as connect indicator,power on.
2.Press ‘SET key’ for 3 seconds,enter into first menu.then the channel measure value window display parameter and the channel setting window display paraneter value,you can adjust these parameters value by poressing ‘▲’、‘▼’、‘◄’、and save the setting by press ‘function key ’,every parameter should be saved after setting ,then enter next parameter set.
3.You can press ‘CH1’、‘CH2’ for 3 seconds to enter into second menu .After finished one parameter set,press ‘SET’ to save it,then enter into next parameter set .
The meter use in the first time or the surroundings have changer, finding it control not good, in this time you need use the setting itself. For example:The first set the AT=1,A-M light flickered, the meter enter into setting itself, press the SET key enter into parameter setting. Set the Hy is 0.5～1℃, if the output is relay set the t=2S, the meter have three times vibrate, automatic preserved P, I, D parameter and the A-M light off, the setting itself finish.
Ⅷ、Fault Analysis and Clearance
XMT*JK208 series adopt advanced production process, and have the strict test before leaving factory, it improve the reliability of the meter .The usual fault caused by the wrong operation or parameter setting .If you find the fault couldn’t be cope with, please record it, and contact with the agent or us. Sheet 7-1 is the usual fault of XMT*JK208 series in the daily application：
Sheet8-1 Common fault disposal
Remark：Our company will improve product technology, design and specification, it is confirm to the object.
Attached 1：Statement of meter’s parameter attention letter and English letter