China Manufacturer for Ac110-220v Temperature Controller - XMT-908P Series Intelligent programmable Temperature Controller – Gongyi
China Manufacturer for Ac110-220v Temperature Controller - XMT-908P Series Intelligent programmable Temperature Controller – Gongyi Detail:
Ⅰ、Main Technical Indexes
1、Basic error： ≤±0.5%F.S±1B
2、Cold end compensating deviation： ≤±2.0℃
3、Sampling period：0.5 second
4、Control cycle：relay output 2～120S , other is 2S.
5、Alarm output the drop in level：0.5 or 5
6、Relay output contact capacity：AC220V/5A (resistance load) or AC220V/0.3A (perceptual load)
7、Dring controllable pulse output：≥3V scope，≥50μS widthcontact with pulse when move to exceed zero
8、Driving solid relay signal output: driving electric current≥15mA, voltage≥9V.
9、Continuous PID : 0～10mA (load 500±200Ω), 4～20mA (load 250±100Ω),
or 0～5V (load≥100kΩ) , 1～5V (load≥100kΩ)
10、Power: AC90V～242V, 50/60Hz
11、Work environment: temperature 0～50.0℃, humidity≤85%RH, without corrode and strong electric radiation.
Ⅱ、Product code:XMT □ 9 □ 8 □ P (1) ⑵ (3) (4) (5)（6）
(1) Meter faceplate and Installation dimension(mm)：
Blank: 160×80×130 156×76； A: 96×96×110 92×92；
D: 72×72×110 68×68； E: 48×96×110 44×92；
F: 96×48×110 92×44； S: 80×160×130 76×156；
G: 48×48×110 44×44
(2)Design serial number
(3)Alarm： “ 0”: no set alarm “ 1or2”: one alarm “5”: voice alarm
“3”: two group alarm (Alarm mode can be selected by the parameter ALP)
(4)Input signal： “8”: Free exchange of the input signal
(5)Control method： Blank: Relay contact; “G”: Solid state relay;
“A”: Zero passed trigger of controlled silicon； “ A3”: Three-phase over-zero trigger adjustment;
“ B”: Controlled silicon phase-shifted trigger； “C”: 0-10mA or 4-20mA constant electric current output;
(6)“P”：64 segment program(It contains 32 rise temperature segments and 32 constant temperature segments.)
Ⅲ、Panel schematic diagram(consult)
Ⅳ、Code setting mode
Time parameter :the third setting area（only use when with the print function）
5.1、The first setting area:
Press the SET key 3S enter into the first setting area, the meter will display the parameter code 1～23 in the window at the upper row and display the parameter data at the low row. In this time press the ▲、▼ or t key to adjust the parameter, then press the SET key to preserve. If within 10 seconds do not press every key then it will automatically to preserve the data and withdraw the setting.
The LOCK is electronics lock, when Lock=0, all the parameter can be revised; when Lock=1, only the “SP” can be revised; when the Lock＞1, all the parameter can not be revised. But don’t set the Lock＞50
5.2、The second setting area (Partial parameter of curve program control):
Curve parameter setting area, press SET+3 Second to enter into the other operation is the same as above.
5.3、The third setting area (Time parameter):
When the power on, press SET＋▲＋▼key to enter into the setting area, the other is the same as above.
5.4、When run=0, The setting value amend(SP) :
Press the ▲ key 3 second to enter into the setting value amend area(SP), the other is the same as above.
When the meter is set up with the electricity, press the t key about 3S enter into the manual regulation, it will display “H” at the lower row, in this time can set the output power; press the t key about 3S again it will withdraw the manual regulation.
When the control object is valve, the manually operation value >50,and is co rotating，whereas is reversal，stable output duty ratio is 100%。
5.6、reposition: press t +▼key about 1 second, the program will reposition the first stage, run according to the running state.
5.7、Normal using ,it shows the measurement data in the window at the upper row and display the setting data SV at the lower row, press the▼ key it will display the main control output data, the first LED display “F”，latter three LED 0～100 output data.
5.8、Suspension and recycling operation :
When the instrument is in the normal operation, pressing ▲ key for 3 seconds program segment will enter into the suspension state, this time AM indicator light is flashing; pressing ▲ key for 3 seconds again, the program segment resume running. When the program segment finish running, then enter standby mode; pressing ▲ key for 3 seconds the program segment start to re-run from the first segment.
Ⅵ、Operation method of running curve program
1、Three work state of the meter
State of stop:when the meter is in the state of stop, the meter is used as the thermostatic control, the setting value is the basic setting value（SP）,display in the window below, the running indicator light turn off.
State of running：
When the meter is in the state of running ,the meter constantly modify the setting value （SV）according to the setting curve，make the measured value（PV） changing according to the curve program, so as to achieve the purpose of curve program control, and the indicator light turn bright.
State of interim ：
When the meter is in the state of interim, the calculagraph stop timing ,the setting value（SV）maintain unchanged, and it also will prolong the running time of the curve program, the indicator light will coruscate.
State of automatic pause：
The state of automatic pause is the special form of the state of interim, created by the meter itself, not can be controlled by artifial. In the running state, when the deviation absolutevalue of themeasured value and present setting value（SV）>automatic pause strap (AL_P)，the meter enter into the pause state，the indicator light will coruscate.，the calculagraph stop timing ,the setting value（SV）will not change. when the deviation absolutevalue of themeasured value and present setting value（SV）< automatic pause strap (AL_P),the meter will automatically resume the running state.
when the meter finish the curve program, and some section r=0时,the meter will enter into the hold state (run=1)， and the main control output closed, the indicator light turn off, the below window display the first setting value. When the outer switch touch off or set the meter Run parameter to 3, it can restart.
2、Disposal on the power-cut
During the running process of the curve program，the meter will for every 5 minutes to save the running parameter and the running state data，so when the power is off and then on ，the meter only run according to the last saved data to continue ，
not from the beginning(if it need to start from beginning, press reposition key to start.
The meter use in the first time or the surroundings have changer, finding it control not good, in this time you need use the setting itself. For example:
Set the HY is 0.5～1℃, if the output is relay set the t=2S, then set the AT=1, A-M(or AT) light flickered, in this time the meter enter into setting itself. It have three times vibrate, automatic preserved P, I, D parameter and the A-M(or AT) light off, the setting itself finish.
Note：①when Setting itself, the instrument should not change the set value.
② When the power off during setting itself, as the meter has the memory, it will restart setting itself next time.
③ When it need artificially exit during setting itself, set the parameter to 0 so that can exit，but the setting result will not be valid.
④The parameter set suitable for most of the system, but not for the minority system. so we can adjust P、I、D value. when artificially adjust, look into the response curve. If it is the short cycle oscillate (about the same long as the oscillation cycle of setting itself or on-off control ), decrease P (priolity), increase I and D; If it is the long cycle oscillate (more times as on-off control), increase I(priolity), increase P、D; if with no oscillate but with steady-state error. decrease I(priolity), increase P; if last can control steady but need long time decrease D(priolity), increase P, decrease I. The adjustment can adopt step-by –step method, first to increase or decrease 30-50% with one parameter of P、I、D. If the control result is get better, then keep on increasing or decreasing the parameter till the result is best. In general, we modify P first, then I, if the result is also not well, and modify D parameter. When modify these three parameter, we should consider the overshoot and control precision these two index.
When output control valve, as the cycle of open and closed is too long, it should artificially modify PID parameter on the basis of Ex-Factory value if the result of setting itself is not well. (In general P↑ on the basis of Ex-Factory value, diminish and in order to avoid continual action, D should adjust smaller.
Note①: Only one way for alarm use Alarm 1.only when alarm method ALP is 3(upper and lower limit alarm)、6(upper and lower deviation alarm)、9(upper and upper limit alarm)、10 (lower and lower limit alarm),it should use Alarm 2,when upper and lower limit alarm or upper and lower deviation alarm, it should use Alarm 2 as lower limit or lower deviation alarm. The indicator light will be lighten when the alarm output.
Note ②: when input current signal 0-10mAor 4-20mA, it should respectively combine 1K or 250 250Ω to input port. Change the current signal to voltage signal.
.As below take the surface of the paste solder equipment for instance setting the program section parameter：
1、 Prewarming zone: Rise the stove temperature from the starting temperature to soaking zone. r1=30℃/minutes；
2、 Soaking zone：namely the first platform section C1=150℃ heat preservation T1=2minute；
3、 Circumfluence zone：the program skip from the first platform to the second platform（r2=200℃/minute）,to make the stove rise to the soldering temperature C2=220℃,heat preservation T2=1minute；
4、 Cooling zone：set r3=199℃/minute，andC3=80℃ to make cooling rapidly. and set T3=0 minute so as to finish the whole jointing process.
5、 Reposition switch to stand by state：After the temperature decrease to 80℃ to enter into the third platform zone, but as T3=0, then enter into the fourth slope zone directly. when r4=0, so the meter skip to the first program zone to enter into the stand by state（pro=1 ； run=1）。
.Temperature Curve chart as below：
Attached 1：Statement of meter’s parameter attention letter and English letter
★Note: Our company will continue to improve product technology, design specification. If change, please subject to the material object, without notice.
Product detail pictures:
Related Product Guide:
We now have a highly efficient crew to deal with inquiries from clients. Our intention is "100% shopper pleasure by our merchandise quality, price tag & our staff service" and take pleasure in a very good standing amongst purchasers. With quite a few factories, we can easily provide a wide vary of China Manufacturer for Ac110-220v Temperature Controller - XMT-908P Series Intelligent programmable Temperature Controller – Gongyi, The product will supply to all over the world, such as: Manchester , Tanzania , Belize , If you need any of our products, or have other items to be produced, please send us your inquiries, samples or detailed drawings. Meanwhile, aiming to develop into an international enterprise group, we look forward to receiving offers for joint ventures and other cooperative projects.
By EliecerJimenez from Swansea - 2018.09.21 11:44
Perfect services, quality products and competitive prices, we have work many times, every time is delighted, wish continue to maintain!
By olivier musset from Malta - 2018.07.12 12:19